Under normal circumstances a connection will be established within a second. Do not yet connect the microscope and peripherals to the power supply at this point! Care of the Microscope 54 6. There may be situations that the default 1. This value should be small, around zero. This will have different firmware to the manual posted by Jason as that was for a Stonex X9. Click this button to display further details. This setting, together with the Scan Resolution, has a huge impact on the amount of data.
The change of the language becomes only effective after restarting the software. You need to be connected in order to select any Action see below. Suggested Remedies Recharge or change battery. Only Leica Geosystems authorised service centres are enti- tled to repair these products. The driver will always communicate sending and receiving commands via this command port number of the scanner. Highly recommended, in order to use exact time-stamping of the laser data. When going online for the very first time, the Windows operating system firewall may come up with a security alert.
Notes on the disposal of the device, accessories and consumable materials. The internal lithium ion batteries allow a duration of 2. Interval 0 for a desired sector. If no network connection can be made at all e. A new, embedded tilt sensor allows averaged, batch-style tilt compensation after each scan that opens up the potential use of convenient traverse and back-sight field methods, in addition to the traditional use of standard target-based registration and geo-referencing methods. In order to maintain this condition and to ensure safe operation, the user must follow the instructions and warnings contained in this operating manual.
Using patched fuses or bridging the fuse holder is not permitted. Especially the Vertical or Scan Area Selection is very important. However, these are not strict rules, because each project is different and depends on the current situation and hardware being used. Controller Computation Setup Disabling the laser system in your Computation Setup will have the most effect on the performance. Notice there is a physical mask from 0°-20° and 340°-360°.
Notice that it may take a few seconds before the first data is received, so be patience for at least 10 seconds. To skip all pixels, set Interval to 0, to use all pixels, set Interval to 1 and to use every n-th pixel only, set Interval greater than 1 Notice that omitting a Sector Interval value means 0 skip all Performance-wise, it is not recommended to use a Sector Reduction scheme to skip all pixels i. When switching on the instrument or starting a scan, the system switches off automatically even though it was totally recharged. Still, we are always grateful for comments and suggestions regarding potential mistakes within this documentation. The instrument is operated Start scanning, for example at a temperature lower perform some preview scans, as than -10°C. Dual-axis compensator and laser plummet laser class 2 are components of the measuring instrument.
To export the scan data and inserted targets of the project, click the export button in the Review Acquisition menu. The antenna has limited mobility. The actual number of beams will be variable, and depends for example on the used on-line settings and on the targets being scanned. This includes the examination of samples taken from the human body with a view to provide information on physiological or pathological states or congenital abnormalities, or to determine the safety and compatibility with potential recipients, or to monitor therapeutic measures. Instead you should use a Mask Scheme of the Vertical Area Selection, see above. At the workplace, large temperature fluctuations, direct sunlight and vibrations should be avoided. Problems If you experience problems using your laser scanner in combination with this driver, or if you need additional information or support, please attach the daily laser log-file when submitting your.
The software calculates the new spatial resolution of the encoders. A higher speed is recommended when the laser unit is mounted on a fast moving vehicle. With nationwide service and fast mobilization, we can quickly respond with customized laser scanning solutions for your project. In the manual you sent me does not specify the export format. Notice that some combinations with the Scan Resolution are not allowed.
Under normal conditions you would expect values of 2000 to 5000 pixels. The laser will also stop scanning automatically when going offline with the Controller. Caution: Before completing the stage, make sure no objectives are installed! Under normal conditions the memory used by these processes should be constant. Product identification The type and the serial number of your product are indicated on the type plate. Replay you may use small cell-sizes, e. The standard thread is M25. When on-line, the laser unit can be controlled using the Controller: Select Echosounder Settings, click on the Laser System icon, and select the 'Control' tab page.
The real-time datastream is received via this port number plus one, so normally 6101. It is important to select the right combination with the Scan Speed. The scanner has a maximum scan range of 187 meters. Note: Like other electronic devices, the microscope and its accessory components must not be disposed of as regular household waste. Changing one parameter changes the other two parameters accordingly. The microscope's electrical accessory components are not protected against water.